Range:0 Hz,100 Hz to maximum frequency of device 0 Hz,100 Hz
Accuracy:± span /?swept points -1?
Reference frequency:10.000000 MHz
Reference frequency accuracy:±(?days from last calibrate x freq aging rate?+ temperature stability + initial accuracy )
Temperature stability:<2.5ppm ?15? to 35??
Marker frequency resolution:span/?the number of sweep points -1?
Uncertainty:±?freq indication x freq reference uncertainty +1%x span +10%x resolution bandwidth + Marker Frequency Resolution? Resolution:1 Hz,10 Hz,100 Hz,1 kHz
Accuracy:±?marker freq x freq reference uncertainty + counter resolution?
Resolution bandwidth (-3 dB):10Hz to 500kHz?in 1 to 10 sequence?, 1MHz,3MHz
Resolution filter shape factor:<5:1
1 x Spectrum Analyzer
1 x power cable
1 x usb cable
1 x CD
1 x manual